Pathway Rectal Sensor 6340
Pelvic Floor Stim-Equipment & Accessories, sEMG Accessories, Sensors, Internal
Pathway Rectal Sensor 6340SKU: PG-6340
Combination sEMG Biofeedback and Pelvic Floor Stimulation (PFS) rectal sensor. The PG-6340 internal sensor was originally designed for rectal use but is commonly used vaginally to accommodate patients with smaller vault sizes.
Web ID: 426
Combination sEMG Biofeedback and Pelvic Floor Stimulation (PFS) rectal sensor. This rectal sensor is designed as a single patient multi use sensor and cannot be used in between patients. The PG-6340 internal sensor was originally designed for rectal use but is commonly used vaginally to accomodate patients with smaller vault sizes.
The rectal sensor has two tabs for insertion identification. The longer of the two tabs should be facing up towards the pubis in order to achieve the desired placement. The sensor features a bulb at the end. The bulb helps the sensor to seat correctly in the orifice and guards against expulsion for those patients who are extremely weak or have a prolapse.
The sensor has three elongated nickel contact points, these contacts act as your two active and one reference electrode during sEMG biofeedback training. The length and width of the contacts provide a larger area of dispersement, allowing the clinician to deliver a more comfortable Pelvic Floor Stimulation to the patient.
As a single patient multi use item, clean up is simple. The recommended cleaning for the sensor is soap and water and let the sensor air dry. For infection control purposes it is recommended that the clinician not store the sensors for individual patients in the clinic. Patients should take their sensors with them and bring them back to the clinic for their next visit.
This sensor will directly connect to the Pathway CTS systems through port A of the device and be used through that port to record sEMG biofeedback and deliver PFS. The sensor will also connect to the Pathway MR and TR series sEMG instrumentation through the use of the PG-3660 adapter. It is designed to directly input into the PG-STM-10 and can be used in conjunction with the liberty PFS unit with the use of the PG-3700 adapter.
This product does not contain any rubber or latex materials.